Abstract: the turbine meter
scale installation note information produced by excellent flow meter, flow meter manufacturer to offer you the quotation.
is widely used in measurement in a closed pipe with 1 cr18ni9ti stainless steel, 2 cr13 and corrode corundum AI2O3, cemented carbide, and no fiber, particles and other impurities, kinematic viscosity is less than 510 - working temperatures
6 m2 / s liquid, the kinematic viscosity than 51.
More flow meter manufacturers choose model price quote you are welcome to inquire, here is a turbine meter scale installation considerations article details.
Turbine flowmeter is widely used in measurement in a closed pipe with 1 cr18ni9ti stainless steel, 2 cr13 and corrode corundum AI2O3, cemented carbide, and no fiber, particles and other impurities, motion viscosity temperature less than 5 & times;
6 m2 / s liquid, the kinematic viscosity is greater than 5 & times;
6 m2 / s of the liquid, can be used for solid liquid flow meter calibration after, you can also control quantitatively, excessive alarm, etc. , is an ideal instrument to flow measurement and energy saving.
When installation considerations: 1 level of turbine flowmeter installation of piping should not have visual sensor can detect skew (
Vertical installation of sensor pipe vertical deviation should be less than 5 & deg;
And flow direction should be the bottom up.
2 can not stop the flow, and globe valves shall be equipped with bypass.
If the fluid contains impurities, should be in the upstream side filter;
If the containing gas, should to the upstream side gets over.
Filters and venting of sewage outlet and venting mouth will lead to a safe place.
3 flow regulating valve should be installed in the downstream of the sensor, the upper reaches of the globe valve should be fully open when measuring, and are not allowed to produce vibration and leaked.
For the likely should add the check valve to prevent reverse flow process of fluid flow.
In 4 sensors should be one of pipe, sealing washer shall not protruding into the line.
Sensor should no longer be horizontal peak, lest accumulate pipeline gas (
Such as stop flow when mixed with air)
Stay in the sensor, not easy eduction, affect the measurement.
5 sensors should support both before and after the line, do not produce vibration.
For easy to condensate fluid to the sensor and the pipeline before and after heat preservation measures.
Above is all there is in this article, you are welcome to inquire me manufacturer flowmeter selection, quotation, etc.
Sincerity is the leading manufacturer of mass flow meter and related products.
Are you interested in buying ? We also have all kinds of in offer. Visit Sincerity Flow Meter to know more and order, we have them at pocket friendly prices.
By investing in an ethical supply chain, Beijing Sincerity Automatic Equipment Co., Ltd position ourselves to engage with a driven, engaged customer base.
Beijing Sincerity Automatic Equipment Co., Ltd will do this by managing our business with integrity and the highest ethical standards, while acting in a socially responsible manner with particular emphasis on the well-being of our teammates and the communities we serve.
give you an additional digital fork density meter option for your u shape coriolis mass flow meter, whether it being a tuning fork liquid density meter, coriolis density meter or endress hauser coriolis mass flow meter. Go and get more info at Sincerity Flow Meter.