Abstract: the installation of rotameter scale information produced by excellent flow meter
, flow meter manufacturer to offer you the quotation.
, generally divided into glass rotor flowmeter and metal pipe float flowmeter, generally under the premise of no maintenance after installation can run normally for a long time, so about the installation of the rotor flow meter, also need attention.
Measurement of cooling.
More flow meter manufacturers choose model price quote you are welcome to inquire, here is a rotameter scale details the installation method of the article.
Rotor flowmeter, generally divided into glass rotor flowmeter and metal pipe float flowmeter, generally under the premise of no maintenance after installation can run normally for a long time, so about the installation of the rotor flow meter, also need attention.
Measuring temperature and easy to precipitate crystallization or easy to solidification of liquid, should choose jacketed metal tube rotor flow meter.
Rotameter installed note: 1, the instrument must be vertical direction most of the rotameter installed on no vibration of the pipeline installation, should not have obvious tilt, fluid flow through the instrument from bottom to top.
Instrument no strict length of straight pipe upstream requirements, but also have manufacturing requirements (
2, used in the installation of the dirty fluid should be in the upstream filter instrument.
With magnetic coupling tube rotor flow meter used in sexual impurity fluid may contain magnet, magnetic filters should be installed in front of the instrument.
To maintain the cleanness of float and cone tube, especially the small instrument, float clean degree significantly affect the measured values.
3, rotor flowmeter flow value should make necessary conversion without according to the use of the density, viscosity and other medium parameters to the rotor flow meter manufacturer specially instrument, liquid instrument calibration to water often flow, gas meter with air calibration, fixed value in engineering standard state.
Conditions of use of fluid density, gas pressure, temperature and calibration are inconsistent, to do the necessary conversion.
Conversion formula and method of rotor flowmeter factory have a detailed instruction manual.
Main rotor flowmeter measurement object is single-phase liquid or gas, particles in the liquid to solid or gas containing droplets usually do not apply.
Because the float attached to particles in the liquid flow or tiny bubbles will affect the measurements.
If need far transmission signal for integrating total or flow control, electrical signal output is generally selected tube rotor flow meter (
The metal flow meter)
Such as environmental atmosphere with explosion-proof requirements and field control instrument air supply, the preferred far eastone metal tube rotameter.
All in all, different using environment and use requirements, choose different types of flowmeter selection is particularly important.
Above is all there is in this article, you are welcome to inquire me manufacturer flowmeter selection, quotation, etc.
As a entrepreneur, being trapped in a company under multiple quality problems never appealed to Beijing Sincerity Automatic Equipment Co., Ltd.
Are you looking for ? Beijing Sincerity Automatic Equipment Co., Ltd has the collection you want, like v shape coriolis mass flow meter or electromagnetic flow meter slurry and many more in the online stores. Visit Sincerity Flow Meter to know more.
Beijing Sincerity Automatic Equipment Co., Ltd provides a way for you to understand your customers, to learn what makes them unique and what motivates their behavior. We can then leverage that wealth of information to personalize our interactions and demonstrate that mass flow meter is valuable to our customers.
coriolis density meter mass flow meter will help keep your tuning fork liquid density meter in a digital fork density meter state.
You will have a insertion ultrasonic flow meter that looks vortex flow meter manufacturer all the time, Because your is handling with it.