Abstract: the turbine meter
scale installation points parsing information produced by excellent flow meter, flow meter manufacturer to offer you the quotation.
Changzhou into abundant flow instrument co. , LTD LWGY basic turbine flow sensor to measure liquid instantaneous flow and cumulative total volume, can also control quantitative liquid.
Sensor has high precision, long service life, simple operation and maintenance and so on, is widely used.
More flow meter manufacturers choose model price quote you are welcome to inquire, here is a turbine meter scale installation points parse article details.
City into abundant flow meter LWGY basic turbine flow sensor to measure liquid instantaneous flow and cumulative total volume, can also control quantitative liquid.
Sensor has high precision, long service life, simple operation and maintenance and other characteristics, widely used in factories, oil field, chemical industry, metallurgy, paper making, is an ideal instrument to flow measurement and energy saving.
* quality, user * & throughout;
, bring more convenience and benefit to the customer.
1, flowmeter installation point of the inner diameter of the upstream and downstream piping is the same as the meter diameter.
2, flowmeter must be installed on the pipeline (
Tilted pipe within 50)
Axis, installation flowmeter should be concentric to the pipe axis, flow direction should be consistent.
Flowmeter upstream pipe length should be not less than 2 d straight pipe section, if the installation site allows advice straight pipe upstream to downstream of 20 d, 5 d.
3, good flowmeter zui installed indoors, must be installed in the outdoor, be sure to use prevent bask in, prevent the rain.
Lightning protection measures, so as not to affect the service life.
4, in order to ensure not affect the normal use of medium maintenance flow, the flow meter should be installed to cut off the valve in the pipeline before and after (
The by-pass pipe shall be set up, and at the same time.
Flow control valve to be installed downstream of the flow meter, flowmeter upstream of the valve must be fully open when using, avoid the upper part of the produced fluid unsteady flow phenomena.
5, flow meter should be installed in the convenience of maintenance, without strong electromagnetic interference and thermal radiation.
6, in order to guarantee the life of the flowmeter, filters should be installed at flow meters in front of the straight pipe.
7, flow meter should be reliable grounding, not common ground with high voltage systems.
8, the user to match a pair of standard flange welding on the pipe before and after.
Not allowed to take meter welding!
Before install flowmeter should strictly eliminate dirt such as welding slag in the pipeline, zui works such as the diameter of the pipe (
Or the by-pass pipe)
Instead of flowmeter for venting pipe.
In order to ensure that in use process flow meter from being damaged.
Install the flow time, the sealing gasket between the flange cannot be sunken into the pipe.
, from factory has been committed to flow meter for decades and the research and development and production of the material level meter, over the years, we sell not only products, but also a kind of attitude and service, pay attention to is the feeling of the customer experience, from the perspective of customers, strive to achieve consistent perfect pre-sale and after-sale!
Above is all there is in this article, you are welcome to inquire me manufacturer flowmeter selection, quotation, etc.
With new and upcoming social commerce technologies, the biggest change for mass flow meter marketers will be a shift in focus from branding to lead generation and conversion.
Many websites provide additional information on the topic of mass flow meter. One such site worth visiting is Sincerity Flow Meter.
Beijing Sincerity Automatic Equipment Co., Ltd expects to reach the desired profits in the first year and does not anticipate serious cash flow problems.
Our company is professional in selling mass flow meter as well as providing a series of relevant services.