Abstract: the selection and application of precession vortex flow meter
scale information produced by excellent flow meter, flow meter manufacturer to offer you the quotation.
In order to better use ZK -
LUX precession vortex flowmeter
, attention should be paid to the following: 1, the correct selection of precession vortex flowmeter selection should grasp the two basic principles in the process, to ensure the production safety, second, we must ensure the accuracy.
To do this.
More flow meter manufacturers choose model price quote you are welcome to inquire, here is the correct selection and application of precession vortex flow meter scale article details.
In order to better use ZK -
LUX precession vortex flowmeter, attention should be paid to the following: 1, the correct selection of precession vortex flowmeter selection should grasp the two basic principles in the process, to ensure the production safety, second, we must ensure the accuracy.
To do this, we must to carry out the selection of the three parameters, namely the immediate and long-term zui, zui small and common instantaneous flow (
Mainly used for the size of the selected instrument specifications)
And the design of the measured medium pressure (
It is mainly used for the selected instrument nominal pressure level)
, the actual work pressure (
Mainly used for the selected instrument of pressure sensor pressure level)
2, before using the calibration in order to ensure that the instrument in the future work in the process of the measurement result is reliable and accurate, should be in formal before installation to be sent to the detection capability and qualification of authorities, conduct a verification system of the whole flow range.
Installed 3, note that although this kind of instrument technology and the special requirements of using the environment doesn't have much, but any kind of flow measurement instrument has some common, such as much as possible to avoid vibration and high temperature environment, away from the flow interference elements (
Such as compressor, separator, pressure regulating valve, reducer and pipe, elbow, etc. )
Straight pipe before and after, to keep the instrument inside sleek straight, guarantee the clean of single-phase fluid measured medium, etc.
4, the late management in order to guarantee the accuracy of the instrument work for a long time, reliability,
To avoid unexpected shutdown and loss of data)
Periodically system calibration (the,
Every 1 ~ 2 years)
, transcribing header data (
Daily or weekly)
, replace the medium parameters (
Monthly or quarterly)
And don't regularly check the battery status, check the instrument coefficient and seal etc.
5, internal temperament smudgy maintenance or other reasons, the need for instrument measuring cavity and its components in regular inspection or cleaning.
One thing must pay special attention to: for the same specifications of the vortex precession flowmeter, the vortex core components such as body, diversion and are not interchangeable.
Otherwise, he must calibration instrument measurement coefficient and the matching of the temperature and pressure sensor calibration system.
Above is all there is in this article, you are welcome to inquire me manufacturer flowmeter selection, quotation, etc.
This is an internationally recognized standard which acts as a form of guarantee that everything Beijing Sincerity Automatic Equipment Co., Ltd does is managed to the highest quality standards.
insertion ultrasonic flow meter are the in thing today. To buy a for yourself do visit Beijing Sincerity Automatic Equipment Co., Ltd at Sincerity Flow Meter.
Beijing Sincerity Automatic Equipment Co., Ltd manufactures mass flow meter with innovative facilities and professional operation.
The group's Quality Systems Manager (QSM) is responsible for ensuring that Beijing Sincerity Automatic Equipment Co., Ltd has in place systems that guarantee quality throughout the Group.
If we avoid overselling mass flow meter coriolis mass flow meter emerson and capabilities, you gain credibility and earn trust. We did not want to jeopardize our ingrained preference for humility.