Abstract: the intelligent electromagnetic flowmeter
scale field calibration method of information produced by excellent flow meter
, flow meter manufacturer to offer you the quotation.
Huaian far can instrument co. , LTD. , in the production of electromagnetic flow meter factory, its accuracy has been is line calibration.
But in the use of the site, due to the environmental conditions, fluid characteristics, as well as the instrument itself such as components damage reason lead to the failure of the instrument operation and so on.
More flow meter manufacturers choose model price quote you are welcome to inquire, here is the intelligent electromagnetic flowmeter scale field calibration method the article details.
Huaian factory in the production of electromagnetic flowmeter, its accuracy has been is line calibration.
But in the use of the site, due to the environmental conditions, fluid characteristics, as well as the instrument itself such as components damage reason lead to the failure of the instrument operation, etc. , to use for a long time after the flow meter is necessary a routine calibration.
As a result of the intelligent electromagnetic flowmeter sensor installed on the pipeline, it is difficult to remove the sent to the calibration line for inspection, and at present within the scope of the flow measurement, national and local on site conditions to large diameter flowmeter has yet to file online inspection procedures and other regulations.
Huaian to ensure electromagnetic flowmeter after troubleshooting and the accuracy and reliability of after use for a long time, and shall not affect the daily production, electromagnetic flowmeter calibration with six kinds of methods: 1, safety of electromagnetic flowmeter excitation coil insulation test, should be more than 20 m & Omega;
2, the excitation coil electromagnetic flowmeter copper resistance test, should be the same as the original factory value (
The environment temperature phase at the same time)
3, resistance to electromagnetic flowmeter sensor electrodes to test, if the resistance value between the 2 a 20 kf, accompanied by charge and discharge phenomenon, the resistance of the two electrodes, is that good.
4, to the test of electromagnetic flow meter transducer excitation current, the output was observed and the converter current value, the error is not more than 0.
5, the electromagnetic flowmeter converter analog output and frequency output test, observe the change of the linear, and calculate its zui linear error, should be less than 0.
6, to the electromagnetic flowmeter, above DN1200mm should test driver stage NB, current error is not more than 12 ma.
Above is all there is in this article, you are welcome to inquire me manufacturer flowmeter selection, quotation, etc.
Beijing Sincerity Automatic Equipment Co., Ltd is dedicated to servicing our customers with a wide array of high-quality service and products.
As a result, consumers will reward Sincerity with leadership sales, profit, and value creation, allowing our customers in which we live and work to prosper.mass flow meter
is attracting a great positive feedback from the customers. And many of our clients are fully satisfied with it.
The development of coriolis density meter mass flow meter products has massive potential for expansion.