Orifice flowmeter in the application
orifice meter application has a long history, has the international standard, the theory of high precision, has been widely applied, but after decades of application, found the orifice flowmeter has the following: (1) application of many factors, Design parameters and operating conditions, lack of straight pipe upstream, different heart, orifice plate and pipe orifice a-side polluted, acute wear, etc. ) Has a very large effect on the accuracy of measurement, the measurement error; (2) the installation of more trouble, maintenance and unpick and wash larger workload. (3) need to be used with a differential pressure transmitter, increased maintenance workload, the other must guide pressure pipe laying, and should be carried out on the pressure pipe heat preservation in winter, can not be installed outdoors; (4) the flow range ratio of 1:3, limitations; 5. If the installation is not correct, prone to leakage; 6. The pressure loss is bigger, the high cost of operation.
Beijing Sincerity Automatic Equipment Co., Ltd's products, whether interim or permanent, comply fully with all appropriate producing regulations.
Beijing Sincerity Automatic Equipment Co., Ltd also maintains a friendly, fair, and creative work environment, which respects diversity, new ideas, and hard work.
While the productivity and efficiency benefits of automation are unequivocal for manufacturing mass flow meter, the need for skilled humans to operate, utilize and advance technologies is equally unmistakable.
Beijing Sincerity Automatic Equipment Co., Ltd is a initial company that supports expertise in searching marketing solutions.
Beijing Sincerity Automatic Equipment Co., Ltd knew the only way to remain competitive was to ensure quality of service and customer satisfaction above all.