Abstract: the measurement errors of the target flowmeter
scale caused the main reason for the large information produced by excellent flow meter, flow meter manufacturer to offer you the quotation.
Target flowmeter measurement error caused the main reason for the big target flowmeter work appear abnormal value increase, in the process of measurement error is large, some of the reasons for this phenomenon, its zui the main reason for the following: a target, the target and stem with filiform.
More flow meter manufacturers choose model price quote you are welcome to inquire, here is the target flowmeter scale measurement error of the main causes of great articles for details.
Target flowmeter measurement error caused the main reason for the big target flowmeter work appear abnormal value increase, in the process of measurement error is large, some of the reasons for this phenomenon, its zui the main reason for the following: a, target and target stem with filamentous and banded sundry;
Methods: reference to deal with sundry method.
B, under the condition of high knot engine-grease, target and target pole have serious engine-grease, target plate along the measured flow stress element, pipe axis projection area increase, namely the target piece to measure flow tube between annular flow area is reduced, and under the same flow rate, the sensor stress increases, zui has led to an increased flow of abnormal value;
Processing method: remove the transition parts, use tool to target and target pole and measuring the flow tube wall engine-grease things clear.
C target flowmeter, installation and connection pipe a larger relative concentricity, sealing gasket is not concentric, thus forming the throttle resistance, greatly influence the measured liquid, gas, dust flow;
Methods: adjust the install state.
D, straight pipe before and after the target flowmeter is too short, and prior to the target flowmeter directly installed the elbow, valve, such as greatly interfere with the measured liquid, gas, dust flow components;
Methods: according to the specifications for installation or real flow field was carried out on the target flowmeter calibration.
E, by-pass pipe leakage;
Methods: inspection and replace the by-pass pipe.
F, target sheet wrapped around have banded sundry, increases the target bearing;
Processing method clutter processing method: with reference to the front.
Magnetic flap inspcetion liquidometer intelligent electromagnetic flowmeter level gauge
stainless steel pressure gauge above the target flowmeter is all there is in this article, you are welcome to inquire me factory flowmeter selection, quotation, etc.
Whether it's automation or artificial intelligence, the rapid convergence of technology and business often determines mass flow meter
Click Sincerity Flow Meter for super quality from one of the state's premier producers.
Data has always been important in business, of course. But with the arrival of digital data—its volume, depth, and accessibility—it has become clear it is key to helping Beijing Sincerity Automatic Equipment Co., Ltd develop sustainable competitive advantage.