Flowmeter fluid flow, flow display is not zero
flowmeter fluid flow, flow display is not zero, or unstable display value cause analysis: 1. Shielding grounding, a transmission line interference signals mixed with display input terminal 2. Pipeline vibration, the impeller jitter, 3 produce false signals. Cut-off valve closed lax caused by leakage, in fact, the instrument shows that leakage of 4. Display the internal circuit board or electronic components between metamorphic damage, interference solution: 1. Check the shielding layer, if they have a good grounding 2 display terminal. Before and after reinforcement line, or in the sensor with 3 stents to prevent vibration. Repair or replace the valve 4. Take & other; Short-circuit method & throughout; Or item by item, one by one check, judge the interference source, find out fault point
Beijing Sincerity Automatic Equipment Co., Ltd guarantees to providing quality products and services.
Helping our customers manage document workflow and increase efficiency through best-in-class mass flow meter and services. Fostering the growth and development of our employees.
We focus on operational procedure and manufacturing facilities of mass flow meter.
Forging an tight connection starts with understanding your potential customers and catering to their needs on mass flow meter, both with a quality product and impactful endress hauser ultrasonic flow meter.