Differential pressure type orifice flow meter and the relationship between the differential pressure
differential pressure type flow and the relationship between the differential pressure of orifice flowmeter, illustrated as follows: pressure transmitter measuring flow, the differential of 0 ~ 100 kpa, secondary table range of 0-200 t/h, differential pressure transmitter output current for 4 & ndash; 20 ma, test for flow of 50 t/h and 160 t/h when the corresponding differential pressure value and current value is how much?
1) According to the differential pressure flow formula:
F = K× △p△px /△p = ( Fmax/F） 2 ( Square) △px =△p ( Fmax/F） 2 = 100000 ( 50/200) 2 ( Square) = 6. 25 kpai = ( Fmax/F） × 16+4=8mA△px=100000( 160/200) 2 ( Square) =64KPaI=( 160/200) × 16 + 4 = 16. 8 mA
2) Flow rate and differential pressure is generally open to relationships,
the F * F = K * delta p, K = F× F/delta P = 200 * 200/100 = 400 kpa < / P> < P> when flow rate of 50 t/h: delta P1 F1 / K = 50 * * F1 = 50/400 = 6. 25 kpa for the output current ( 20mA- 4mA） × 6. 25/100 + 4 = 5 ma
flow for 160 t/h: delta P2 = F2 * F2 / K = 160 * 160/400 = 64 kpa for the output current ( 20mA- 4mA） × 64/100 + 4 = 14. 24马
Beijing Sincerity Automatic Equipment Co., Ltd has created a professional team which contained with a numbers of engineers and technology experts.
We are proud to be one of the largest suppliers out there. You will love what we offer for your mass flow meter solution. Check our website at Sincerity Flow Meter or call to talk to our customer service department with any questions you may have.
mass flow meter has a very good repute over the global market.
Regularly improving mass flow meter in accordance with customer feedback is a great way to show your brand listens and cares.
The development of coriolis density meter mass flow meter products has massive potential for expansion.